CLINITEST® hCG Test on CLINITEK Analyzers - Rapid and Reliable Pregnancy TestingFebruary 15th, 2018
Study: Exposure to Air affects Integrity of Urine Reagent StripsFebruary 14th, 2018
AACC Annual Scientific Meeting and Clinical Lab Expo
Medlab - The World’s Largest Expo
DubaiBooth No.: Z5 G 42
MEDICA 2017 - World Forum for Medicine
Düsseldorf, GermanyBooth No.: 3/D35-2
69th AACC Annual Meeting and Clinical Lab Expo - July 30 - August 3, 2017
SAN DIEGO, CA, USA
Thanks to Block Scientific, I was able to procure the re-certified Bayer DCA 2000+ without hassles and get the lab back in operation. The
device works perfectly and I look forward to doing more business with Block Scientific.
--- Mathew Anderson, New Jersey
Enzymatic Potassium Test Kit (liquid stable), R1: 4 x 40 mL, R2: 4 x 10 mL
The assay shows virtually no affect by the following interfering substances at indicated concentrations: Na+ 150mM, NH4+ 0.5 mM, Ca2+ 7.5 mM, Pi 2.0 mM, ascorbic acid 10.0 mM, Zn2+ 0.5 mM, Fe3+ 0.5 mM, Cu2+ 0.5 mM, triglycerides 1000 mg/dL, hemoglobin 500 mg/dL, conjugated bilirubin 20 mg/dL and unconjugated bilirubin 15 mg/dL. The assay has a linearity range from 2-8 mM. Within precision %CVs are 1.12% and 1.20% for 4.46 mM and 6.86 mM potassium serum samples, respectively. Total precision %CVs are 1.77% and 1.77% for 4.46 mM and 6.86 mM potassium serum samples, respectively. Comparison study showed that the assay has good correlation with existing ISE method with a correlation coefficient of 0.98.
For in vitro quantitative determination of potassium in fresh human serum. Measurements obtained by this assay are used to monitor electrolyte balance in the diagnosis and treatment of diseases conditions characterized by low or high blood potassium levels. For IVD use only.